With the CONELOG® abutments, customized esthetic bridge restorations can be fabricated.
A screw-retained temporary bridge restoration is performed with temporary abutments made of titanium alloy of the bridge version. To compensate implant axis divergences, the abutments are not equipped with a tapered implant-abutment connection. The abutments can be veneered directly with plastic.
The prefabricated CONELOG® Vario SR, prosthetic components precisely matched to one another can be used to fabricate occlusally screw-retained crown, bridge and bar constructions. 20° and 30° angled CONELOG® Vario SR abutments are well suited for bridging large implant axes divergences. Where bone volme is reduced and anatomical structures are unfavorable for implantation, the implants can be placed in the distal direction and an appropriate prosthetic restoration can be created.
CONELOG® Titanium bases CAD/CAM of the bridge version act as bonding base for customized bridge constructions made from suitable materials. Digital recording of the implant or lab analog positions is performed with CONELOG® Scanbodies. Fabrication of the bridge construction is performed with CAD technology. The geometries of the titanium bases are filed in a CAD library at www.camlog.com under Digital Technology. To compensate implant axis divergences in case of primary splinting, the titanium bases CAD/CAM for bridges are not equipped with a Tube-in-Tube® Implant-abutment connection.